Transcranial magnetic stimulation (TMS) is a method in which a changing magnetic field is used to cause electric current to flow in a small region of the brain via electromagnetic induction. During a TMS procedure, a magnetic field generator, or “coil”, is placed near the head of the person receiving the treatment.:3 The coil is connected to a pulse generator, or stimulator, that delivers a changing electric current to the coil.
TMS is used diagnostically to measure the connection between the brain and a muscle to evaluate damage from stroke, multiple sclerosis, amyotrophic lateral sclerosis, movement disorders, motor neuron disease, and injuries and other disorders affecting the facial and other cranial nerves and the spinal cord.
Evidence suggests it is useful for neuropathic pain and treatment-resistant major depressive disorder. A 2015 Cochrane review found that there was not enough evidence to determine its effectiveness in treating schizophrenia. For negative symptoms another review found possible efficacy. As of 2014, all other investigated uses of repetitive TMS have only possible or no clinical efficacy.
Matching the discomfort of TMS to distinguish true effects from placebo is an important and challenging issue that influences the results of clinical trials. The greatest risks of TMS are the rare occurrence of syncope (fainting) and even less commonly, induced seizures. Other adverse effects of TMS include discomfort or pain, transient induction of hypomania, transient cognitive changes, transient hearing loss, and induced currents in electrical circuits in implanted devices.
The use of TMS can be divided into diagnostic and therapeutic uses.
TMS can be used clinically to measure activity and function of specific brain circuits in humans. The most robust and widely accepted use is in measuring the connection between the primary motor cortex and a muscle to evaluate damage from stroke, multiple sclerosis, amyotrophic lateral sclerosis, movement disorders, motor neuron disease and injuries and other disorders affecting the facial and other cranial nerves and the spinal cord. TMS has been suggested as a means of assessing short-interval intracortical inhibition (SICI) which measures the internal pathways of the motor cortex but this use has not yet been validated.
For neuropathic pain, for which there is little effective treatment, high-frequency (HF) repetitive TMS (rTMS) appears effective. For treatment-resistant major depressive disorder, HF-rTMS of the left dorsolateral prefrontal cortex (DLPFC) appears effective and low-frequency (LF) rTMS of the right DLPFC has probable efficacy. The Royal Australia and New Zealand College of Psychiatrists has endorsed rTMS for treatment resistant MDD. As of October 2008, the US Food and Drug Administration authorized the use of rTMS as an effective treatment for clinical depression.
Although TMS is generally regarded as safe, risks increase for therapeutic rTMS compared to single or paired TMS for diagnostic purposes. In the field of therapeutic TMS, risks increase with higher frequencies.
The greatest immediate risk is the rare occurrence of syncope (fainting) and even less commonly, induced seizures.
Other adverse short-term effects of TMS include discomfort or pain, transient induction of hypomania, transient cognitive changes, transient hearing loss, transient impairment of working memory, and induced currents in electrical circuits in implanted devices.
During a transcranial magnetic stimulation (TMS) procedure, a magnetic field generator, or “coil” is placed near the head of the person receiving the treatment.:3 The coil produces small electric currents in the region of the brain just under the coil via electromagnetic induction. The coil is positioned by finding anatomical landmarks on the skull including, but not limited to, the inion or the nasion. The coil is connected to a pulse generator, or stimulator, that delivers electric current to the coil.
Most devices provide a shallow magnetic field that affects neurons mostly on the surface of the brain, delivered with coil shaped like the number eight. Some devices can provide magnetic fields that can penetrate deeper, are used for “deep TMS”, and have different types of coils including the H-coil the C-core coil, and the circular crown coil; as of 2013 the H coil used in devices made by Brainsway were the most developed.
Nexstim obtained 510(k) FDA clearance for the assessment of the primary motor cortex for pre-procedural planning in December 2009 and for neurosurgical planning in June 2011.
A number of deep TMS have received FDA 510k clearance to market for use in adults with treatment resistant major depressive disorders.
The use of single-pulse TMS was approved by the FDA for treatment of migraines in December 2013. It is approved as a Class II medical device under the “de novo pathway”.
In the European Economic Area, various versions of Deep TMS H-coils has CE marking for Alzheimer’s disease, autism, bipolar disorder, epilepsy  chronic pain major depressive disorder Parkinson’s disease, posttraumatic stress disorder (PTSD),
In 2013, several commercial health insurance plans in the United States, including Anthem, Health Net, and Blue Cross Blue Shield of Nebraska and of Rhode Island, covered TMS for the treatment of depression for the first time. In contrast, UnitedHealthcare issued a medical policy for TMS in 2013 that stated there is insufficient evidence that the procedure is beneficial for health outcomes in patients with depression. UnitedHealthcare noted that methodological concerns raised about the scientific evidence studying TMS for depression include small sample size, lack of a validated sham comparison in randomized controlled studies, and variable uses of outcome measures. Other commercial insurance plans whose 2013 medical coverage policies stated that the role of TMS in the treatment of depression and other disorders had not been clearly established or remained investigational included Aetna, Cigna and Regence.
Policies for Medicare coverage vary among local jurisdictions within the Medicare system, and Medicare coverage for TMS has varied among jurisdictions and with time. For example:
The United Kingdom’s National Institute for Health and Care Excellence (NICE) issues guidance to the National Health Service (NHS) in England, Wales, Scotland and Northern Ireland. NICE guidance does not cover whether or not the NHS should fund a procedure. Local NHS bodies (primary care trusts and hospital trusts) make decisions about funding after considering the clinical effectiveness of the procedure and whether the procedure represents value for money for the NHS.
NICE evaluated TMS for severe depression (IPG 242) in 2007, and subsequently considered TMS for reassessment in January 2011 but did not change its evaluation. The Institute found that TMS is safe, but there is insufficient evidence for its efficacy.
In January 2014, NICE reported the results of an evaluation of TMS for treating and preventing migraine (IPG 477). NICE found that short-term TMS is safe but there is insufficient evidence to evaluate safety for long-term and frequent uses. It found that evidence on the efficacy of TMS for the treatment of migraine is limited in quantity, that evidence for the prevention of migraine is limited in both quality and quantity.
TMS uses electromagnetic induction to generate an electric current across the scalp and skull without physical contact. A plastic-enclosed coil of wire is held next to the skull and when activated, produces a magnetic field oriented orthogonally to the plane of the coil. The magnetic field passes unimpeded through the skin and skull, inducing an oppositely directed current in the brain that activates nearby nerve cells in much the same way as currents applied directly to the cortical surface.
The path of this current is difficult to model because the brain is irregularly shaped and electricity and magnetism are not conducted uniformly throughout its tissues. The magnetic field is about the same strength as an MRI, and the pulse generally reaches no more than 5 centimeters into the brain unless using the deep transcranial magnetic stimulation variant of TMS. Deep TMS can reach up to 6cm into the brain to stimulate deeper layers of the motor cortex, such as that which controls leg motion.
From the BiotSavart law
it has been shown that a current through a wire generates a magnetic field around that wire. Transcranial magnetic stimulation is achieved by quickly discharging current from a large capacitor into a coil to produce pulsed magnetic fields between 2 and 3 T. By directing the magnetic field pulse at a targeted area of the brain, one can either depolarize or hyperpolarize neurons in the brain. The magnetic flux density pulse generated by the current pulse through the coil causes an electric field as explained by the Maxwell-Faraday equation,
This electric field causes a change in the transmembrane current of the neuron, which leads to the depolarization or hyperpolarization of the neuron and the firing of an action potential.
The exact details of how TMS functions are still being explored. The effects of TMS can be divided into two types depending on the mode of stimulation:
MRI images, recorded during TMS of the motor cortex of the brain, have been found to match very closely with PET produced by voluntary movements of the hand muscles innervated by TMS, to 522mm of accuracy. The localisation of motor areas with TMS has also been seen to correlate closely to MEG and also fMRI.
The design of transcranial magnetic stimulation coils used in either treatment or diagnostic/experimental studies may differ in a variety of ways. These differences should be considered in the interpretation of any study result, and the type of coil used should be specified in the study methods for any published reports.
The most important considerations include:
With regard to coil composition, the core material may be either a magnetically inert substrate (i.e., the so-called ‘air-core’ coil design), or possess a solid, ferromagnetically active material (i.e., the so-called ‘solid-core’ design). Solid core coil design result in a more efficient transfer of electrical energy into a magnetic field, with a substantially reduced amount of energy dissipated as heat, and so can be operated under more aggressive duty cycles often mandated in therapeutic protocols, without treatment interruption due to heat accumulation, or the use of an accessory method of cooling the coil during operation. Varying the geometric shape of the coil itself may also result in variations in the focality, shape, and depth of cortical penetration of the magnetic field. Differences in the coil substance as well as the electronic operation of the power supply to the coil may also result in variations in the biophysical characteristics of the resulting magnetic pulse (e.g., width or duration of the magnetic field pulse). All of these features should be considered when comparing results obtained from different studies, with respect to both safety and efficacy.
A number of different types of coils exist, each of which produce different magnetic field patterns. Some examples:
Design variations in the shape of the TMS coils allow much deeper penetration of the brain than the standard depth of 1.52.5cm. Circular crown coils, Hesed (or H-core) coils, double cone coils, and other experimental variations can induce excitation or inhibition of neurons deeper in the brain including activation of motor neurons for the cerebellum, legs and pelvic floor. Though able to penetrate deeper in the brain, they are less able to produce a focused, localized response and are relatively non-focal.
Luigi Galvani did pioneering research on the effects of electricity on the body in the late 1700s, and laid the foundations for the field of electrophysiology. In the 1800s Michael Faraday discovered that an electrical current had a corresponding magnetic field, and that changing one, could change the other. Work to directly stimulate the human brain with electricity started in the late 1800s, and by the 1930s electroconvulsive therapy has been developed by Italian physicians Cerletti and Bini. ECT became widely used to treat mental illness and became overused as it began to be seen as a “psychiatric panacea”, and a backlash against it grew in the 1970s. Around that time Anthony T. Barker began exploring use of magnetic fields to alter electrical signalling in the brain, and the first stable TMS devices were developed around 1985. They were originally intended as diagnostic and research devices, and only later were therapeutic uses explored. The first TMS devices were approved by the FDA in October 2008.
TMS research in animal studies is limited due to early FDA approval of TMS treatment of drug-resistant depression. Because of this, there has been no specific coils for animal models. Hence, there are limited number of TMS coils that can be used for animal studies. There are some attempts in the literature showing new coil designs for mice with an improved stimulation profile.
Areas of research include:
It is difficult to establish a convincing form of “sham” TMS to test for placebo effects during controlled trials in conscious individuals, due to the neck pain, headache and twitching in the scalp or upper face associated with the intervention. “Sham” TMS manipulations can affect cerebral glucose metabolism and MEPs, which may confound results. This problem is exacerbated when using subjective measures of improvement. Placebo responses in trials of rTMS in major depression are negatively associated with refractoriness to treatment, vary among studies and can influence results.
A 2011 review found that only 13.5% of 96 randomized control studies of rTMS to the dorsolateral prefrontal cortex had reported blinding success and that, in those studies, people in real rTMS groups were significantly more likely to think that they had received real TMS, compared with those in sham rTMS groups. Depending on the research question asked and the experimental design, matching the discomfort of rTMS to distinguish true effects from placebo can be an important and challenging issue.
The rest is here:
Transcranial magnetic stimulation – Wikipedia